Multi Energy CT (MECT) - Proposal
1. Proposed Work Item: Multi Energy CT (MECT)
- Proposal Editor: Steven Nichols (email@example.com)
- Proposal Contributors:
- Profile Editor: Steven Nichols (firstname.lastname@example.org)
- Profile Contributors: DICOM WG-21
- Domain: Radiology
Multi-energy CT relies on proprietary private attributes and lacks vendor-independent platforms with a standardized approach to image display. Multi-energy datasets are reconstructed and viewed separately; making it time consuming for the reviewer to page through energy levels for a given anatomic location. DICOM has developed vendor-independent CT modules for encoding enhanced and classic multi-energy images, as well as a blending presentation state that can be used to save and reproduce color overlay images. A multi-energy CT profile could standardize elements of image encoding and display to reduce reliance on proprietary workstations and applications for routine multi-energy review. DICOM workgroup 21 developed initial informational content in Supplement 188 intended to standardize multi-energy CT workflow, however this was not formalized in the final draft. Workgroup 21 remains interested in continuing this work and is willing to participate in the development of an IHE profile. The IHE process has the potential to engage users and vendors to establish baseline functionality for multi-energy display. This can be used to establish upstream image encoding requirements. The additional ability to expand upon underlying multi-energy concepts within a profile can help educate image display vendors on critical elements of multi-energy CT display and workflow.
2. The Problem
Commercial multi-energy CT solutions were established prior to the development of Supplement 188. These relied on a combination of private attributes and dedicated display applications to address user needs for image display. As multi-energy CT has become increasingly more routine, demand has also increased to present multi-energy CT studies on general radiology PACS displays. This profile will help make multi-energy display features more commonly available.
3. Key Use Case
A patient undergoes a multi-energy CT study in which various energy data sets are reconstructed. All data sets are networked to the image manager/archive. Relevant prior CT studies are prefetched for comparison. Comparison studies are potentially monoenergetic or multi-energy, from a different vendor or model CT scanner, and may be from a different institution. The interpreting radiologist requires that the current and prior studies be displayed with the additional features required to properly display multi-energy objective, material quantification and material visualization images without resorting to the need for an acquisition-vendor-specific proprietary workstation. When interrogating the multi-energy images, the radiologist may examine the same image at various energies, apply color overlay to differentiate materials (e.g. to demonstrate separation of uric acid and calcifications) and apply a ROI or point measurement reporting the real-world value of a pathologic feature. A presentation state may be created to save the color palate, blending and annotations applied during image review. A few months later, at an external facility, the virtual monoenergetic images are imported into a legacy display system that supports the Enhanced CT SOP class, however this system does not recognize the study as multi-energy. The images are properly labeled, providing an indication to the reviewer that CT pixel values in virtual monoenergetic images are different from those in conventional CT images, avoiding confusion or misinterpretation.
4. Standards & Systems
DICOM CT Image Storage SOP Class DICOM Enhanced CT SOP Class DICOM Blending Softcopy Presentation State SOP Class
5. Technical Approach
- Acquisition Modality
- Image Display
- Modality Images Stored [RAD-8]
- DICOM CT Image Storage SOP Class
- DICOM Enhanced CT SOP Class
- Retrieve Images [RAD-16]
- Add attributes displayed and display functionality
- Blending Presentation States Stored [RAD-57]
Content to be developed
- Define viewing features required and reach user / vendor consensus.
- Identify any additional DICOM attributes for display.
- Identify dimensions for multi-frame objects.
Impact on existing integration profiles
- Establish a minimum set of Image Display viewing features required by users.
- Reaching a vendor consensus for multi-frame dimensions.
6. Support & Resources
- DICOM Workgroup 21
- User participation to develop a minimum set of display features and establish a boundary between image display and post processing. If there are several users involved, reaching a consensus could pose a challenge.
- If a vendor consensus cannot be reached for multi-frame dimensions due to equipment differences, or prospective vs retrospective image reconstruction workflow, this can be included in the profile as a concept section.
8. Tech Cmte Evaluation
<The technical committee will use this area to record details of the effort estimation, etc.>
Effort Evaluation (as a % of Tech Cmte Bandwidth):
- xx% for MUE
- yy% for MUE + optional
<Delete this Category Templates line since your specific Profile Proposal page is no longer a template.>